HEPA filter Airborne Particle Counter for Compressed Air
Airborne Particle Counter (APC) is fully compliant with ISO14644-2015 and ISO21501-4 standard.
Capacitive touch screen8.0″
Data 5 million sets
8 inch capacitive touchscreen, quick response
Even with two pairs of latex gloves
Sealed 316L stainless steel enclosure and dust plug to prohibit dust and water
Automatically Collect Air Samples
APC data exceed the limit,triggers the BAS to start sampling
-Standard report: one key to generate ISO/EUGMPI CHINESEGMP reports
-Data review and data export
1. Sum up data by zone.
2.Freely switch counting unit between △/Σ/#,m /ft.
3.Automatically generate UCL values and average values
4.Search for sample records precisely by time,zone and operator.
-Data integrity: featured with audit trail.
- Real-time concentration curves, to get a better understanding of environmental particle condition.
- Quick editing function, to make sampling more flexible.
- Clear layout for orders. Simple and easy to use.
- One-click for system software upgrade.
- Multiple sets of personalized sampling schemes
can be customized for any zone.
Device is used for displaying the information of the software and firmware. It includes device name, model, serial number, hardware version, software version, etc. The software can be upgraded on this interface. See chart 5-3-15.
Update is used for upgrading the software version. APC supports continuous version upgrades and cross version upgrades. Insert the MicronView USB flash drive into the USB A port at the back of the device. Touch the【Update】icon, search and select the upgrade file, touch【√】icon to confirm. See chart 5-3-16.
Back- up is used to export all information in the device to the MicronView USB flash drive in the form of db files. When connected to the USB flash drive, touch【Back-up】icon, and a prompt box ‘Backup completed’ will display after the back-up is complete. Touch【OK】icon to confirm. Restore is used to import device information in the form of db file to the device. Insert the MicronView USB flash drive with the db file includes device information into the USB A port at the back of the device. Touch【Restore】icon to restore the device information. When restoring, it is necessary to confirm that the serial number and version of the db file name in the USB flash drive are the same as those displayed in the device information; otherwise, the prompt message of ‘restoration failure’ will be displayed. After the restoration is completed, the device automatically shut down. After restarting, the device can be used normally.